NI-VISA not installed in the systematall NI-VISA is required if you want to communicate through the serial port using LabVIEW. VI_ERROR_INV_MECH BFFF0027h Invalid mechanism specified. Timeout expired before operation completed, shown in the following dialog: Why do I receive this error and how do I resolve it? This tells you how many bytes that command sends back.Note: If you still receive the error while reading only 1 byte, verify the command to make sure it has been written Source
The device will continuously output the data whether the program is on or off. Only one program can use the serial port at a time. Poor|Excellent Yes No Document Quality? You can still run another VI using serial from LabVIEW, but not an executable.
Click Yes. For VisaNS properties, you can also work around this issue by using the SetAttributeInt32 method to assign a value to the attribute. In this mode, a carriage return is \r, a line feed is \n, and an end of line is \r\n. Answered Your Question? 1 2 3 4 5 Document needs work?
The system returned: (22) Invalid argument The remote host or network may be down. Primary Software: Driver Software>>NI-VISA Primary Software Version: 4.7 Primary Software Fixed Version: N/A Secondary Software: LabVIEW Development Systems, LabWindows/CVI Development Systems Hardware: Instrument Connectivity Other, Serial, VXI/VME/MXI, GPIB Problem: When I Make sure you have LabVIEW, NI Serial, and NI VISA on your computer. Labview Error 200279 For more information on how resource names are formatted, see the VISA Resource Name Control help.
Applications designed to interface with this type of instrumentation should anticipate this delay, and implement a Wait function between VISA Write and Read to effectively avoid a potential timeout error. Right-click on Software. VI_ERROR_INV_ACCESS_KEY BFFF0021h The access key to the resource associated with this session is invalid. http://digital.ni.com/public.nsf/allkb/A2561CDE98B66DD4862566510078264F Note: These solutions also help resolve Error 37: Device Not Found that occurs when using Serial Compatibility VIs.
Note: Changing the Timeout value for your VISA session will affect all VISA Writes and Reads in your session. Labview Visa Read Error VI_ERROR_INV_OFFSET BFFF0051h Invalid offset specified. The file serpdrv is not available (only for LabVIEW 6.1 and earlier Serial Compatibility VIs). My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation.
Note: NI-VISA can be obtained from the National InstrumentsDrivers page. http://digital.ni.com/public.nsf/allkb/874B379E24C0A0D686256FCF007A6EA0 Primary Software: Driver Software>>NI-VISA Primary Software Version: 4.5.1 Primary Software Fixed Version: 5.3 Secondary Software: Measurement Studio>>Measurement Studio for .NET Support Problem: I am running an application that uses the VisaNS Labview Error Code -1073807339 VI_ERROR_INV_CONTEXT BFFF002Ah Specified event context is invalid. Labview Error 1073807339 If you are running an executable built in LabVIEW and LabVIEW is open, LabVIEW may already have control of the serial port.
More data might be available. 1073676300 VISA received more event information of the specified type than the configured queue size could hold. 1073676407 The specified configuration either does not exist or this contact form Poor|Excellent Yes No Document Quality? VI_ERROR_INV_LINE BFFF00A0h The value specified by the line parameter is invalid. VISA-specified defaults will be used. 1073676413 Session opened successfully, but the device at the specified address is not responding. 1073676414 The path from trigSrc to trigDest is already mapped. 1073676416 Wait Labview Error Daqmx
Refer to the KnowledgeBase for more information about correcting errors in LabVIEW. VI_ERROR_INV_PARAMETER BFFF0078h The value of some parameter--which parameter is not known--is invalid. Why does this happen? have a peek here A framing error occurs when a byte of data is received, but it is not in the format expected.
Verify that the command being sent to the serial device has the termination character that the device requires. Error Labview Fpga My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation. Then, by using a VISA Flush I/O Buffer VI, you can clear any invalid data in the buffer.
Primary Software: LabVIEW Development Systems>>LabVIEW Full Development System Primary Software Version: 1.0 Primary Software Fixed Version: N/A Secondary Software: N/A Problem: I am outputting a sequence of characters from a third Why am I getting this error and how can I eliminate it? Please Contact NI for all product and support inquiries. Labview Fpga Error Code VI_ERROR_INV_SIZE BFFF007Bh Invalid size of window specified.
VI_ERROR_INV_ACC_MODE BFFF0013h Invalid access mode. Poor|Excellent Yes No Document Quality? To verify if NI-VISA is installed, open Measurement & Automation Explorer, expand the "Software" section in the left column, and verify that NI-VISA appears in the list. Check This Out VI_ERROR_NSUP_INTR BFFF009Fh The interface cannot generate an interrupt on the requested level or with the requested statusID value.
VI_ERROR_NPERMISSION BFFF00A8h Access to the resource or remote machine is denied. Solution: This error is a framing error. Visa32.dll missingThis DLL should be located at WINDOWS\system32. VI_ERROR_INTR_PENDING BFFF0068h An interrupt is still pending from a previous call.
Missing Run-Time Engine: If you are deploying this application, verify that your deployment machine has the NI-VISA Run-Time Engine installed. VI_ERROR_INV_RSRC_NAME BFFF0012h Invalid resource reference specified. Installing NI-VISA and NI-Serial to a Real-Time System: Open Measurement and Automation Explorer Navigate to Remote Systems » Your Device's Hostname or IP. Please tell us why.
VI_ERROR_TMO BFFF0015h Timeout expired before operation completed. Poor|Excellent Yes No Document Quality? VI_ERROR_WINDOW_MAPPED BFFF0080h The specified session currently contains a mapped window. Answered Your Question? 1 2 3 4 5 Document needs work?
VI_ERROR_MEM_NSHARED BFFF009Dh The device does not export any memory. Please Contact NI for all product and support inquiries. VI_ERROR_QUEUE_OVERFLOW BFFF002Dh The event queue for the specified type has overflowed (usually due to previous events not having been closed). VI_ERROR_MACHINE_NAVAIL BFFF00A7h The remote machine does not exist or is not accepting any connections.
VI_ERROR_RAW_RD_PROT_VIOL BFFF0035h Violation of raw read protocol occurred during transfer.